DJO/Chattanooga no longer supports or repairs these units
AFI still supports all Intelect Legend units and is fully stocked
with every replacement part, component and accessory to fix your
2 or 4 channel muscle stimulator/EMS, ultrasound only or
combination with ultrasound
Some of the more common problems include...
- E04, HHH, unable to read calibration data and one wire
device errors are most often a problem with the applicator
not the machine - before servicing try a different
applicator - if that solves the problem the faulty
applicator can be repaired, exchanged or replaced.
- stim board reset error, stim board failure, overcurrent
error, and unit freezing at testing stim board message tend
to be stim board problems that require repairing or
replacing the board by a trained technician.
- faulty power supplies can cause erratic behavior - the
unit might power on but with no display, beep constantly, or
have display issues - if another power supply is available
try replacing it before calling for service
- if some buttons stop working or get stuck but others are
OK the overlay probably needs to be replaced - especially if
the faulty buttons are the more commonly used ones like
intensity up, time up, start or stop - this repair requires
service by a trained technician.
If you think your unit needs service by a trained technician,
please give us a call.
you should return How to
return instruments How long
repairs will take Loan instruments
great rental programs that can meet all your needs.
Trade in your not
working or old equipment for credit toward any equipment sale,
rental or service! Any condition is OK. We are always looking for
scrap parts and units. AFI
Medical is one of the oldest and most trusted equipment service
companies in the New England area. We are the preferred repair
specialists for many manufacturers and have earned the trust of
thousands of customers.